File(s) stored on our OPeNDAP server
Please note: Linked content is NOT stored on 4TU.ResearchData and we can't guarantee its availability, quality, security or accept any liability.
Measured data from sensor Q3 at Maisbich for month 2004-10
datasetposted on 08.10.2009, 00:00 by W.M.J.(Wim) Luxemburg, Cees-Anton van den Dool
Parent item: Maisbich (Luxemburg), measured data from discharge meter Q3, 2003-12 to 2006-10 Measuring instrument: Maisbich basin: V-notch discharge meter Q3 (Qdown) Type: Pressure Sensor Keller DCX-4.0, SN 1467 This instrument is based on differential pressure measurement of two absolute sensors. The sensors measure pressure and temperature. Height of V-notch above sensor: -0.0015m. Datasets with correction contain the corrected discharge observed at the Vnotch. The discharge is calculated as Q=Cd*h^2.5 After correcting the logger measurements with hand measured depths we interpolated the difference (dH) Then we determined the factor Cd with discharge measurements which we also interpolated The timing of hand measurements can be corrected to better represent the logged values The last corrections (variable "clean_corr") were made for blocking of the V-notch weir as folows: z1 represents the last good discharge value after which sedimentation slowly starts blocking the weir z2 represents the moment when the blocking is max z3 represents the moment just before cleaning of the weir z4 represents the moment after cleaning of the weir The max difference in discharge is Q(z3)-Q(z4)